Intelligent servo tank gauge is a combination of site conditions of the petroleum, chemical, metallurgy, medicine, food and other industries, developed with independent intellectual property rights of a new generation of intelligent level measurement instruments. Because of this instrument in the measurement method and the traditional methods of measurement makes it totally different from the measurement accuracy is the world’s all level measurement instrument can not be compared. Intelligent servo tank gauge can be installed through the top of the tank meets the requirements of a single or multiple measurements. The servo tank gauge with my plant tank intelligent management system supporting the use of tanks in the region can be for all tank level, temperature, pressure and other centralized measurement, control, display and print reports, and according to the above the implementation of automated metering tank parameters and data upload, thus truly a “convenience”, “intuitive.” Modernized tank farm of intelligent management.
Scope: versatile measurement and standard output signals as well as the integration of lightweight design makes Intelligent servo tank gauge can be widely used in various areas to be measured.
Chemical industry: in the field of chemical industry, a variety of anti-corrosion material shell of arbitrary choice, to meet the chemical industry in different media level, interface and other measurement requirements.
Food industry: in the beverage industry, a large number of liquid products throughout the production process, an accurate measurement of the liquid level is directly related to the cost of production control and production management.
Power industry: fuel, accurate measurement of oil and large boiler water level is directly related to the production of safe and accurate management.
How it works:
Servo tank gauge measurement is based on the principle of buoyancy balance and design. Fixed within the magnet wheel motor drive with precision-machined drum (ie outside the magnet wheel) to complete the magnetic coupling between. So that the drum and the drive motor is formed synchronously rotated, the meter housing is completely isolated from the magnet between the inner wheel and the drum (outer magnet wheel), so that the measured liquid chamber and the electrical part is completely isolated and meet proof requirements, uniform measurement wire neatly arranged on the drum. The float was sent to the tank by measuring the wire. When the tank level (interface or specific gravity) changes due to the weight of the float level varies with changes, changes in the weight of the float makes the drum (outer magnet wheel) is formed between the inner magnet wheel and coupling poor. After the signal is sent to the microprocessor to calculate judge to issue control instructions to the servo motor, making the tank with the liquid level of the float always changes.
This article comes from tradeindia edit released